Total amount: € 0,00
Official Journal of the , , , ,
In association with
Indexed/Abstracted in: CINAHL, Current Contents/Clinical Medicine, EMBASE, PubMed/MEDLINE, Science Citation Index Expanded (SciSearch), Scopus
Impact Factor 2,063
Online ISSN 1973-9095
Ali S., Osman N. A., Razak A., Hussain S., Wan Abas W. A. B.
Department of Biomedical Engineering, Faculty of Engineering, University of Malaya, Kuala Lumpur, Malaysia
BACKGROUND: Lower limb amputee’s are greatly affected in dealing with the environmental barriers such as ramps and stairs and reported high interface pressure between the residual limb and socket/liner. Interface pressure between the residual limb and socket/liner can affect the satisfaction and use of the prosthesis. Until now, little attention has been paid to interface pressure between socket and stump during ramp negotiation and its effect on amputee’s satisfaction.
AIM: The aim of this study was to evaluate the interface pressure produced by two different liners (Seal-In X5 and Dermo) between the residual limb and socket, and their effects on amputee’s satisfaction during ramp negotiation.
DESIGN: Observational study.
SETTING: The study was performed in rehabilitation and biomedical departments of University Malaya Medical Centre.
POPULATION: Total ten (7 male, 3 female) transtibial amputees with unilateral amputation were included.
METHODS: Two prostheses were fabricated for each amputee. After four weeks of acclimation period, interface pressure between socket and residual limb was measured during walking on ramp and Prosthetic Evaluation Questionnaire (PEQ) was filled for each liner.
RESULTS: Mean peak pressure was significantly (P<0.05) lower with the Dermo liner compared with the Seal-In X5 liner in ramp walking. In addition, the participants were more satisfied with the Dermo liner (83.50 vs. 71.50) and mentioned fewer problems (87.00 vs. 69.00) compared with the Seal-In X5 liner during ramp negotiation.
CONCLUSION: It might be concluded that Dermo liner could be a good choice for the transtibial level of amputation due to relative decrease in interface pressure, satisfaction and fewer problems.
CLINICAL REHABILITATION IMPACT: The advantages of the Dermo liner may improve clinical rehabilitation of transtibial amputee’s, as it provides more satisfaction and experienced fewer problems during ramp negotiation. This provides an improved walking and better quality of life in long term.