Total amount: € 0,00
Indexed/Abstracted in: BIOSIS Previews, Current Contents/Clinical Medicine, EMBASE, PubMed/MEDLINE, Science Citation Index Expanded (SciSearch), Scopus
Impact Factor 1,632
Online ISSN 1827-191X
Sezai A., Shiono M., Akiyama K., Orime Y., Hata H., Yagi S., Tsukamoto S., Nakata K., Hata M., Negishi N., Sezai Y.
From the Second Department of Surgery Nihon University School of Medicine, Tokyo, Japan
Background. Although this examination for mitral and aortic prosthetic valves has been frequently reported to be available for the early diagnosis of complications, there are few reports on examinations for tricuspid prosthetic valves, particularly the mechanical valve such as the St. Jude Medical (SJM) valve. Furthermore, the criteria for early diagnosis of complications are unknown. The purpose of this study is to evaluate the SJM valve in the tricuspid position by Doppler echocardiography and to analyze valve-related complications.
Methods. Between September 1978 and March 1997, 46 patients underwent tricuspid valve replacements with SJM valves; these cases were evaluated by Doppler echocardiography. We divided the cases into Normal and Complication groups.
Results. In the Normal group, peak pressure gradient was 4.2±1.5 mmHg and pressure half-time was 123.5±22.9 msec. In the Complication group, peak pressure gradient was 12.7±3.6 mmHg and pressure half-time was 271.8±76.4 msec. In the Complication group, the lowest peak pressure gradient case was 8.0 mmHg and the shortest pressure half-time was 160.0 msec.
Conclusions. We concluded that when the peak pressure gradient is more than 8 mmHg or the pressure half-time is more than 160 msec, there is a possibility of complications. Despite the problems, the absence of any structural change after more than ten years suggests that the SJM valve can be used in the tricuspid position with careful follow-up.